Mold incubator is generally used in research and application fields such as medical and health, bio-pharmaceutical, agricultural research, environmental protection, etc. It is a special equipment for water body analysis, BOD measurement, cultivation, preservation of bacteria, strains, microorganisms, plant cultivation, and breeding experiments. .
1. Adjust the height with the bottom adjustment screw of the mold incubator to make the box stable.
2. Plug in the power socket (the power supply should be well grounded), press the power switch, the display is bright, and the display shows the actual temperature and humidity in the incubator room.
3. Installation of the humidifier: Plug the power plug of the humidifier into the power socket on the back of the instrument, and connect the humidification tube of the instrument to the humidifier. The connection must be tightly connected. The water in the humidifier tank must be properly operated according to the instructions.
4. Temperature adjustment: Press the temperature setting button, the digital display is the set value, rotate the temperature to adjust the potentiometer to the desired temperature value, release the button, the digital display is the actual temperature in the culture chamber. At this time, if the actual temperature in the incubator is smaller than the set temperature, the heating indicator light is on, and the heater starts to heat; if the actual temperature in the incubator is larger than the set temperature, the cooling indicator light is on, and the refrigeration system starts to cool; If the cooling indicator is dark, the incubator is at a constant temperature.
5. Humidity adjustment: press the humidity setting button, the digital display is the set value, rotate the humidity to adjust the potentiometer to the desired humidity value, release the button, the digital display is the actual humidity in the mold incubator. When the actual humidity in the culture chamber is less than the set humidity, the humidification humidifies the culture chamber at this time, and the humidification indicator light is on; when the actual humidity in the culture chamber is greater than the set humidity value, the humidifier stops working at this time, and the humidification indicator light Off.
6. If there is no need to sterilize in the box, the sterilization switch on the panel should be placed in the â€œoffâ€ position.
7. When the temperature is set, the temperature control knob can not be rotated back and forth many times, so as to avoid frequent compressor start-up, causing overload phenomenon of the compressor and affecting the service life of the compressor.
8. There are two sets of fuse boxes on the back of the machine, 2A? is the cooling and heating load fuse box, 8A? is the control power fuse box. If the machine runs faulty, such as temperature control failure, no heating or no cooling, the power must be cut off. Check that the fuses are intact and check the corresponding parts.
9. Care must be taken when handling. The angle between the water and the plane should not be less than 45Â°.
Shanghai Xinzhuang Instrument Co., Ltd. specializes in R&D, manufacturing and sales of cryogenic thermostat products. It adopts high-quality refrigeration compressor, temperature sensor and stirring motor, precise and stable microcomputer control system, precise temperature control and excellent performance. Free delivery throughout the country. Reasonable price, quality three guarantees, lifetime maintenance, to provide customers with fast, professional, high-quality systematic services, welcome your inquiry to buy!
Customized Golf Head Cap,Golf Animal Cap Set,Popular Golf Head Cover,Knitted Animal Headcover
Grand Dragon Sports Company Limited , https://www.golfyy.com